Equipment for Troubleshooting:
To properly test your supply, you need a standard portable digital multimeter (DMM),a power-SWR meter, and a dummy load.
Multimeter: It measures AC/DC voltages, amperage, and ohm resistance of pots and pickups. It is used to measure line voltage and, when set as an ohm meter, it can be used to check the coax cable.
Power-SWR Meter: Our generators have built-in power meters that show forward and reverse power, but for setting up and testing power supplies, an external meter is useful.
Dummy Load: Dummy loads are necessary for troubleshooting the generator at medium and high power levels. Commercial dummy loads are available with a 50-ohm resistive, noninductive impedance, with power levels ranging from a few hundred to many thousands of watts. The dummy load is usually connected to the output of the RF generator with a short 50-ohm coaxial cable and allows you to put maximum power into a standard load while making tests on the generator. When testing RF generators using impedance-matching networks, a dummy load that more accurately represents the load in the system is highly desirable.
Oscilloscope: This is used to observe the signals from the crystal oscillator through the various driver, buffer, and power amplifier stages. The scope will show any signal distortion, noise, or other characteristics that cannot be detected with a voltmeter.
Frequency Counter: The primary use of this piece of equipment is to check the operational frequency of the RF generator and to verify that the crystal oscillator is on frequency. RF generators are regulated by the FCC and must operate on specific frequencies within a given tolerance range. If the actual frequency deviates too much, adjustments may be necessary. Frequency adjustments are madeonly after the equipment has warmed up for at least an hour—it’s important that the generator be at a stable normal temperature before adjusting the frequency.
Spectrum Analyzer: This Oscilloscope-like equipment is useful in diagnosing unusual troubles in a generator that may be related to distortion and harmonics or spurious signals. Most spectrum analyzers are used in RF communications systems where they can show the sidebands developed during modulation as well as harmonic distortion and for detecting and isolating interference.
RFI-EMI Tester: This test equipment is a highly sensitive radio receiver designed to detect stray RF radiation, including harmonics and intermodulation products. The more sophisticated test sets incorporate a spectrum analyzer.
Field Strength Meter: This is a simple radio receiver used to measure the amount of signal radiated from an RF source.
Troubleshooting Safety Tips:
- Remove all jewelry before working on RF generators and related equipment. Metal jewelry can cause serious harm or even death when you are working on high-voltage RF equipment.
- To minimize accidentally touching a voltage source when working inside the equipment on power supplies or high-voltage connections, use only one hand and keep your other hand in your pocket.
- Do not disable any equipment interlocks. With power removed, you can usually troubleshoot in safety. The low voltage (24 V) safety interlock circuit is designed to disable the unit in the event of an interlock fault condition. Low voltage power for the interlock circuit is supplied by a step down transformer located inside the unit. This transformer is designed to provide safe low- voltage operation and provide isolation from the main AC line.
- To avoid an accidental shock, keep your clothes and work area free of water or moisture.
- To prevent electrical shock and/or RF burns, never operate equipment with the covers removed. Never operate without an appropriate cable connected between the RF output connector on the rear panel and the load.
- Equipment must be bonded to Protective Earth (safety ground) prior to operating the unit. Safety ground connection must be made at the unit’s rear panel designated 1/4"- 20 threaded ground stud. The ground wire should be a #14 awg or equivalent (minimum) green/yellow lead.
- Prior to performing system maintenance, repair or other service operations, the generator must be locked out and tagged out to prevent accidentally energizing the system. First disconnect AC input power to the generator, then mount a suitable “Clamshell” type lockout device to the AC input plug such as a Hubbell # HLD2 or equivalent. Follow all manufacturers’ directions for the lockout device.
Troubleshooting First Steps:
WARNING: Equipment must be installed, operated and serviced only by trained, qualified persons.
CAUTION: Breaking the seal or removing the warranty decal from any unit will void the warranty. If internal damage is suspected, contact factory for assistance.
- If a problem with the generator has been identified, verify it by attempting to duplicate the problem.
- Inspect cables and connectors.
- Check for the correct settings of front panel controls, observe any meter readings or display readouts, and re-enter commands from equipment connected to a personal computer or terminal. For instance, equal forward power and reflected power might indicate an open coax cable between the generator and matching unit, or a bad matching unit.
- One of the most obvious problems is simply lack of power to the equipment. Check circuit breakers and reset, if necessary. Verify that there are no bad fuses. If a fuse was blown, replace it with one of the same value and type. Then recheck the equipment.
- If the system is still malfunctioning, in most cases, the next step should be to replace the generator with a functioning spare to get the system back in operation.
If your equipment is still under warranty, the best course at this point is to request an RMA number and return the unit to the factory, or you can visit our contact us page for your nearest sales office.