IEC55011/IEC55022 Limits for conducted and radiated emissions
IEC61000-3-2 Limits for harmonic current emissions
IEC61000-3-3 Limitation of voltage changes,voltage variations and flicker
IEC61000-4-2 Electrostatic discharge immunity
IEC61000-4-3 Radiated immunity
IEC61000-4-4 Electrical fast transient immunity
IEC61000-4-5 Electrical surge immunity
IEC61000-4-6 Conducted disturbance immunity
IEC61000-4-8 Immunity to power frequency magnetic field
IEC61000-4-11 Immunity to voltage dips, short interruptions and variations on supply line
Insulation test voltages based on 250 VAC working voltage.
MOP = Means of Protection.
MOOP = Means of operation protection.
MOPP = Means of patient protection
Earth leakage = Current flowing in the earth conductor
Enclosure leakage = Current flowing to earth via the patient from the enclosure.
Patient leakage = Current flowing to earth via the patient from an applied part.
Patient auxiliary = Current flowing between two applied parts
NC = Normal Conditions SFC = Single Fault conditions
*Patient care equipment maximum earth and enclosure leakage current for the US is 300 μA
Figures quoted are for portable equipment.